|Download Technical Bulletin|
All Solinst Leveloggers use a lithium battery as their power source. Levelogger battery life estimates are based on standard operating conditions, with no external or internal interferences (i.e. under ideal conditions).
The estimates are based on a set sampling rate. For example, the Levelogger 5 has a battery life estimate of ten years, based on a sampling rate of one reading per minute. Below is a table showing standard battery life estimates for each Levelogger type.
A more rapid sampling rate will deplete the battery more quickly. For example, if a Levelogger 5 is set in Continuous Mode (loops and overwrites data when memory is full) at a sampling rate of 1 second, the battery will be drained in about 4 months. If a Levelogger 5 Junior is used at a sampling rate of 5 seconds, the battery will be depleted in approximately 2.5 months.
It is also important to note, in Event Based sampling, although the Levelogger may only log readings in memory when an “Event” occurs, the battery consumption is still mainly a function of sampling rate. This means that a smaller sampling interval (checks for an Event occurrence) will consume battery power more quickly whether readings are stored or not.
What Else Can Affect Levelogger Battery Life?
In addition to sampling rate, there are other internal and external factors that can have an impact on battery life. Leveloggers have a specified operating temperature range; the Levelogger 5 has an operating range of -20ºC to +80ºC. Exposure the temperature extremes can adversely affect the battery life.
Note: Loggers should always be stored or installed with the top hanger cap on (if not using a Direct Read Cable) to prevent unnecessary battery drainage and protect the optical eyes.
There could also be an internal electronic component failure or an issue during a firmware upgrade, which could cause a short, or the Levelogger system to get stuck in an operational loop. This may lead to battery drain.
Moisture on the optical eyes of a Levelogger can also be a problem during communication. Additional battery consumption will occur during any communication process, if condensation, moisture, or dirt is present on the optical eyes. Moisture and dirt impedes communication, and requires more power than would normally be required with a dry, clean connection.
Levelogger Battery Status
The battery % consumption display on the Datalogger Settings window of the Levelogger Software, changes based on usage, type of activity, and the power consumption of the main internal components of the Levelogger.
You may be able to diagnose what has caused your battery to drain using the Levelogger Software Diagnostic Utility. The Utility will display the temperatures your Levelogger has been exposed to, an estimate of the number of logs that were taken, as well as the battery voltage and charge levels.
Dealing with a Drained Battery
Leveloggers can be returned to Solinst for assessment and/or battery replacement. If a Levelogger is within its warranty period, and there is no obvious misuse of the Levelogger, the repairs may be covered under warranty.