See Section 8.2.2 for installing the well cap base and the Support Hanger Bracket for securing the cable when deployed.
Remove the storage caps from the connectors on the Wellhead and Vented Cable.
Ensure you save all the storage caps for future use.
For diagnostic purposes, you may wish to install the end of the Vented Cable with the serial number, to the Wellhead connection at surface
Line up the alignment socket in the Vented Cable connector with the alignment pin in the Wellhead connector.
Carefully push the connectors together. Twist the cable connector slightly until you will feel a small click when the properly aligned connection is made.
Do not twist the Vented Cable or stainless steel cable connector body when connecting to a Wellhead or AquaVent logger. Only twist the stainless steel coupling.
Tighten the stainless steel coupling while holding the stainless steel cable connector body still.
Tighten the stainless steel coupling until it seats. Do not twist the Vented Cable or Wellhead.
Figure 1-10 Connecting a Vented Cable to the AquaVent Wellhead