Ensure the Levelogger is operating with the latest firmware, and that you are using the latest software. Visit the Solinst website or software update feature for assistance.
- Connect the Levelogger to the communications device and start the software.
- Select the appropriate COM or USB port for the connected communications device from the centre drop-down menu.
- Click the ‘Retrieve Settings from Levelogger’ icon. This will retrieve and display the current programmed settings for the connected Levelogger.
- You can now customize the Levelogger including your Project ID, Location, Sampling Mode and Rate, and Future Stop and Start times.
If a number of Leveloggers are to be programmed with identical inputs, clicking the 'Save Default Settings' icon will create a template.
Datalogger Settings Window
Clicking on the embedded in the software will provide you with a short explanation of that feature, e.g. Com Port, Slate Mode, Time Synchronization, etc.
The Rainfall Calibration Constant 'value' of the tipping bucket rain gauge used with the Rainlogger is required when programming the Rainlogger. Consult the Levelogger User Guide for more information on programming the Rainlogger Edge.
Before deploying your LTC Levelogger Edge, be sure to calibrate the instrument. To begin calibration, open the 'Conductivity Cal' tab and follow the steps provided. Consult the Levelogger User Guide for more information.